Evaluation of serological lateral flow assays for severe acute respiratory syndrome coronavirus-2
BMC Infectious Diseases | June 16, 2021
Trombetta BA, Kandigian SE, Kitchen RR, Grauwet K, Webb PK, Miller GA, Jennings CG, Jain S, Miller S, Kuo Y, Sweeney T, Gilboa T, Norman M, Simmons DP, Ramirez CE, Bedard M, Fink C, Ko J, De León Peralta EJ, Watts G, Gomez-Rivas E, Davis V, Barilla RM, Wang J, Cunin P, Bates S, Morrison-Smith C, Nicholson B, Wong E, El-Mufti L, Kann M, Bolling A, Fortin B, Ventresca H, Zhou W, Pardo S, Kwock M, Hazra A, Cheng L, Ahmad QR, Toombs JA, Larson R, Pleskow H, Luo NM, Samaha C, Pandya UM, De Silva P, Zhou S, Ganhadeiro Z, Yohannes S, Gay R, Slavik J, Mukerji SS, Jarolim P, Walt DR, Carlyle BC, Ritterhouse LL and Suliman S
BMC Infect Dis. 2021;21:580
This study was performed using a Simoa Homebrew assay.
COVID-19 has resulted in significant morbidity and mortality worldwide. Lateral flow assays can detect anti-Severe Acute Respiratory Syndrome Coronavirus-2 (SARS-CoV-2) antibodies to monitor transmission. However, standardized evaluation of their accuracy and tools to aid in interpreting results are needed.
We evaluated 20 IgG and IgM assays selected from available tests in April 2020. We evaluated the assays’ performance using 56 pre-pandemic negative and 56 SARS-CoV-2-positive plasma samples, collected 10–40 days after symptom onset, confirmed by a molecular test and analyzed by an ultra-sensitive immunoassay. Finally, we developed a user-friendly web app to extrapolate the positive predictive values based on their accuracy and local prevalence.
Combined IgG + IgM sensitivities ranged from 33.9 to 94.6%, while combined specificities ranged from 92.6 to 100%. The highest sensitivities were detected in Lumiquick for IgG (98.2%), BioHit for both IgM (96.4%), and combined IgG + IgM sensitivity (94.6%). Furthermore, 11 LFAs and 8 LFAs showed perfect specificity for IgG and IgM, respectively, with 15 LFAs showing perfect combined IgG + IgM specificity. Lumiquick had the lowest estimated limit-of-detection (LOD) (0.1 μg/mL), followed by a similar LOD of 1.5 μg/mL for CareHealth, Cellex, KHB, and Vivachek.
We provide a public resource of the accuracy of select lateral flow assays with potential for home testing. The cost-effectiveness, scalable manufacturing process, and suitability for self-testing makes LFAs an attractive option for monitoring disease prevalence and assessing vaccine responsiveness. Our web tool provides an easy-to-use interface to demonstrate the impact of prevalence and test accuracy on the positive predictive values.